Order in Print
SECTION I. IDENTIFICATION AND DESCRIPTION
SECTION III. CALIBRATION PROCESS
Figure 1. Thermal sight (TI) with top cover of case and shroud removed.
Figure 3. Large cover plate removed (top view).
Figure 5. Equipment setup.
Figure 6. Thermocouple to heater plate setup.
Figure 7. Loosening target selector switch mount. (ON/ALIGN switch is shown for clarity only).
Temperature Check - Continued
Figure 9. Example of Dip Switch correction setting for 0.025 to 0.075 C.
APPENDIX A. BAR Target Temperature Worksheet
© Copyright Integrated Publishing, Inc.. All Rights Reserved. Design by